Q - Switched Pigmentation Removal Machine 1064nm / 532nm Wavelength 1 - 5HZ
Contact Person : Zhang
Phone Number : 86-18510160891
|Minimum Order Quantity :||1set||Price :||Negotiable|
|Packaging Details :||international standard 73X50X39CMX1PC||Delivery Time :||1-2 weeks|
|Payment Terms :||L/C, T/T, Western Union||Supply Ability :||10 sets per week|
|Place of Origin:||Beijing, China (Mainland)||Brand Name:||sincoheren,Monaliza|
|Certification:||CE,Medical CE,FDA,TGA,TUV,CFDA||Model Number:||M4C-II|
|Carbon Gel:||Yes||Light Source:||Q-switched Nd: YAG Solid-state Laser|
|Fluence:||≤ 12mJ/cm2||Pulse Duration:||8ns|
nd yag laser tattoo removal machine,
q switched nd yag laser for pigmentation
China beijing sincoheren q switched nd yag laser tattoo removal pigmentation removal machine
How it works?
Therapy of epidermis and dermis pigmentation-----Utilizing the explosive effect of the Nd: YAG laser, the laser permeate the epidermis into the dermis which includes amount of pigment mass. Since the laser pulses in nanosecond but with super high energy, the shot pigment mass swell quickly and breaks into small pieces, which will be eliminated through metabolic system.
Therapy of the dilatation of capillary vessel---- Utilizing the heat effect of the Nd: Yag Laser, the laser is absorbed by the hemachrome in the capillary vessel, then the capillary vessel become blocked and disappear.
First fill the tank with distilled water or pure water (mineral water is not permitted) to 4/5 of the tank, which can be observed from the window in the profile of the console. After that if the screen no display, pls add a little more water till it arrive the tip. But don‟t drown out. Turn on the system, and when the water level descends, refill it again to 4/5 of the tank.
Change the non-hole stopper with the hole stopper.
The adoption of 1064nm transformer or 532nm transformer is decided by the treatment.
Warning:Any operation not followed instructions of the manual may cause radiation hazard. Caution:The smoke and chipping produced during the course of treatment may be biological grain.
Usage of the transformer with wavelength of 532nm
Take off the transformer with wavelength of 1064nm and change with the transformer with wavelength of 532nm. Revolve the transformer suitably and make the standard lines on the transformer and the handle on the same position to get perfect output.
Caution: The light with wavelength of 532nm is harmful to skin. Please read Annex 2 before operation.
|Light source||Q-switched Nd: YAG solid-state laser|
|Frequecy||1, 2, 3, 4, 5, 6Hz|
|Cooling system||Built-in distilled water-circle system|
|Power requirement||220VAC/10A or 110VAC/10A|