Hair Removal Portable Diode 808 Machine Picosecond
Contact Person : Zhang
Phone Number : 86-18510160891
|Minimum Order Quantity :||1set||Price :||Negotiable|
|Packaging Details :||Standard protective package 57X67X134CMX1PC||Delivery Time :||1-2 weeks|
|Payment Terms :||L/C, T/T, Western Union||Supply Ability :||500 Unit/Units per Month|
|Place of Origin:||Beijing, China (Mainland)||Brand Name:||sincoheren,Razorlase|
|Certification:||CE,Medical CE,FDA,TGA,TUV,CFDA||Model Number:||SDL-B,SDL-C|
|Name:||Laser Hair Removal Professional Machines||Cooling System:||Air Cooling+water Cooling+semiconductor Cooling|
|Energy Density:||Max 120J/cm2||Net Weight:||40kg|
Clinical medical high performance diode laser 808nm laser for hair removal machine price salon equipment
The diode laser system emits laser of 808nm wavelength . It is absorbed by the melanin located in the hair follicles , so as to destroy the hair follicle.The hair shafte also absorb the energy from laser,so as to be destroyed and gasified.The the unwanted hair is removed completely and permanently.
● Hair removal
● Permanent hair reduction
● The gold standard in permanent hair reduction——clinically documented and proven results.
● Safe and effective treatments for unwanted hair on all skin types(I-VI)including tanned skin
● Unique output control enables treating nearly and kind of hair
● Advanced UniChill handpiece ensure patients’ safety and comfort
Smart compact design
● Hand piece identification automatically
● Ergonomic design , easy to operate
● Effective&safe procedure
● Recommended treatment settings
Before treatment, please confirm the following information:
1) Make sure that water level is in the stipulated level;
2) Examine whether all accessories have been connected;
3) Make sure that all people in treatment area are wearing protective goggles oreyeshade;
4) Make sure that the emergency switch is releasing.
2 System Start-up
After finishing preparatory work, turn the power inlet switch to the position of “—”. Thenturn the key switch clockwise to the position of “︱”. The system will start-up and the display screen will show “Welcome”.
After System Start-up, the system will load into a self-testingprocess. If there is any error, it will automatically release alarm and the error code will also be shown on the screen. Please take “malfunctions diagnosis and analysis” for reference, or call us or our agents for help immediately.
3. Skin Test
Skin test is very important before treatment; relevant details are as followed:
1) Choose appropriate area for a test patch. If the target treatment area is the whole face, try to choose a position that will not influence facial beauty, such as bottom of ear. If the target treatment area is chest, neck or other parts of the body, then the area that will not affect visual appearance should be chosen for experimental test. Meanwhile, injured or infected skin should not be treated.
2) Before skin test, the surface of skin should be cleaned. A thin layer about 2-3mm of conductive gel should be spread in the skin test area. Take out laser treatment head and closely touch skin vertically with appropriate pressure slide in the direction of hair growth for 2 to 3 times. Wait for 5 to 10 minutes and then remove the gel, observe the test effect.
If there appears erythema or edema around hair follicle without epidermis reactions, it achieves a perfect effect. The following treatment could be set with the same energy and pulse type.
If there is not any visible change on the skin, skin test should be continued with 1-2J/cm2 energy increased.
If the patient feels severe pain or there appears blister in the skin test area, it is probably because the energy is too high or too much repetitive treatment is given. Then the energy should be lowered if the treatment continues.
|Laser Type||Diode Laser|
|Pulse width||5-400ms adjustable|
|Cooling System||Sapphire contact cooling, Macro channel Cooling,Water and Air Cooling|
|Display||10.4 inch color touch LCD screen|
|Energy Density||Max 120J/cm|
|Lifetime||500 millions shots(120J/cm)|