Shr Opt Elight 480nm Ipl Laser Hair Removal Machine
Contact Person : Zhang
Phone Number : 86-18510160891
|Minimum Order Quantity :||1set||Price :||Negotiable|
|Packaging Details :||packaged by standard wooden case for double handled beauty equipment SHR IPL machine 62X68X127CMX1PC 20X31X54CMX1PC||Delivery Time :||1-2 weeks|
|Payment Terms :||L/C, T/T, Western Union||Supply Ability :||10 set per week|
|Place of Origin:||Beijing, China (Mainland)||Brand Name:||Monaliza,PreciPulse|
|Certification:||CE, FDA, TGA, CFDA||Model Number:||SMQ-NYC-3,SMQ-NYC-2,SMQ-NYC,SMQ-NK|
|Style:||Stationary||Light Source:||Intense Pulsed Light|
|Transfer Systerm:||Crystal Light Guide||Pulse Duration:||IPL:up To 150ms SHR:1-10ms|
professional ipl machine,
ipl skin rejuvenation machine
Professional beauty equipment SHR FP IPL 3 mode in 1 face rejuvenation hair removal ipl beauty machine
In accordance with patients’ practical situation, choose relevant indications and treatment Handpiece. After adjusting appropriate parameters setting based on steps shown in 5.2, choose a suitable skin patch for skin test (for detailed content, see 5.5). Before operation, apply a thin layer of cooling gel at treatment area; put head light closely on target skin. Press manual switch on Handpiece and start treatment. It will release a ―di‖ sound each time it shots. Meanwhile, Handpiece counter and screen will respectively count once. After treatment, establish treatment file for each client so as to check right parameters for next time treatment.
Notice: Lifetime of treatment Handpiece: standard lifetime of each treatment Handpiece is 20,000 shots. Energy will be lowered if Handpiece reaches its lifetime. It is suggested to change for a new Handpiece because Handpiece with more than 20000 shot are not in warranty.
1. Carefully protect patients’ eyes during treatment process
2. Second scanning is allowed on diseased area
3. Light crystal should be tightly put on treatment area without applying pressure
4. Leave no blank during treatment and 1/3 overlapping is permitted.
5. Do not shoot light at upper eyelid
6. Direction of treatment head can be changed so as to guarantee a complete treatment
Skin Test (Essential)
Skin test is essential before optical treatment. Details are as follows:
1. If treatment on face is needed, please select skin under ears as test patch, where it is not influencing facial beauty. If treatment areas are in chest, neck or other parts, please select an area not influencing beauty as test patch.
2. Please clean test patch after it is chosen. Apply a thin layer of cooling gel at test patch. Take out treatment Handpiece and make it contacted with treatment area sufficiently. Trigger 2 to 3 slots and 1/3 area of spot size overlapping treatment area is permitted. Wait for 2 to 3 minutes observe treatment area. If light reddening (not swelling) appears to test patch, it means energy chosen is appropriate and operator can choose this energy level and parameter to continue treatment. If there is no visible changes at the patch test area, it means energy is too low, 1－2J/cm2 energy should be added each time. If serious reddening immediately appears to test patch after shot. It means energy is too high, please continue test with lower energy level.
Equipment Shut down
Turn off Handpiece switch after treatment and put it back to Handpiece holder. Touch ―Standby‖, turn the key switch to ―OFF‖ and device will be automatically shut down. Clean light crystal with gauze for next time usage.
Go with the unit
|690-1200nm||Permanently remove unwanted hair on whole body and for all hair colors|
Go with the unit
|560-1200nm||Reduce and remove pigmented lesions e.g. freckles, age spots, sunburn spots; reduce wrinkles; shrink pores, rejuvenate skin and enhance skin elasticity|