Q - Switched Pigmentation Removal Machine 1064nm / 532nm Wavelength 1 - 5HZ
Contact Person : Zhang
Phone Number : 86-18510160891
|Minimum Order Quantity :||1set||Price :||Negotiable|
|Packaging Details :||Internation standard package||Delivery Time :||1-3 weeks|
|Payment Terms :||L/C, T/T, Western Union||Supply Ability :||10 sets per week|
|Place of Origin:||Beijing, China (Mainland)||Brand Name:||sincoheren, Monaliza ,ExQ-Laser|
|Certification:||CE,Medical CE,FDA,TGA,TUV,CFDA||Model Number:||Monaliza-2|
|Pulse Energy:||500mj For 1064nm; 200mj For 532nm||Pulse Energy Range:||100-500mj For 1064nm; 100-200mj For 532nm|
|Frequency:||1-5HZ||Spot Size:||2-10mm Adjustable|
|Aiming Beam:||635nm, <5mW||Electrical Requirements:||110-240VAC 50-60Hz 1200VA|
nd yag laser tattoo removal machine,
q switched nd yag laser for pigmentation
1064/532 nm Nd yag Laser pigment removal TUV ISO certificat machine with moving head Laser pigmentation removal machine
Major components of Q-Switched Nd:YAGLaser Therapy Systems:
——Laser power supply
——Display and control systems
The device adopts a Q-switched Nd:YAGlaser module with bi-wavelength of 1064nm&532nm. the wavelength could be switched by electrical control. For 1064nm,the maximum pulse energy is over 600mJ; For 532nm,the maximum pulse energy is over 300mJ.
The device adopts seven-joint articulated arm, which can be installed to the main console through quickly with locking screws. High reflectivity mirrors of 1064nm and 532nm are adopted in the articulated arm. The output efficiency of the articulated arm is above 80%. A handpiece with adjustable spot size can be installed at the end of the articulated arm. The treatment spots size can be adjusted from 2 to 10mm.
Laser Power Supply
Laser power is used to provide power for laser module to produce pulse laser. The system adopts integrated laser power and the laser power could be communicated with the systemcontrolling core,whichcan control the output voltages and relevant synchronization information and then control the laser moduleenergy output.
The system is includedcirculating cooling system and the cooling medium shall bedeionized water orpurified water. Water pump carries water cycling into the laser device. After laser is cooled down, the water will run through the condensers, then back into water tank. The condenser use fans to cooling the water. In cooling system, water flow sensor and temperature sensor, are used to monitor the working status of cooling system and feedback the related parameters to the controlsystem synchronously, which will control the cooling system in real time.
Display of Control System
Power-on and power-off the system is realized through the powerbutton. press down thepowerbutton in front, the system will be powered on. Once Input the authorized password, the system screen will display relevant parameters, such as system name, version number, self-test information and operation/setting interface. In operation interface, the laser output energy could be adjusted, and the system working status could be controlled. If the system has any problems, the screen will pop out relevant alarming and error message with acousto-optic indication. The emergency switch is used to cut-off the laser output in case of laser accident and protecting the personnel from being injured.
1. Dimensions (without articulated arm, Width×Length×Height): 370 mm ×957 mm ×992 mm；
2. Total weight(include articulated arm): <80kg;
3. Power input: 1200VA;
4. Rated current of leakage:10A;
5. Repetition Rate: 1-5Hz;
6. Nominal ocular hazard distance: NOHD 3.3km,
Note: operation shall be in a closed room and make sure to wear goggles!
7. cooling system: built-in closed circulating water cooling;
1. Laser output mode: Q-switched pulse;
2. Laser wavelength: 1064/532nm；
3. Pulse Duration: 5ns±1ns;
4. Maximum pulse energy in the end of articulated arm: 500mJ@1064nm；200mJ@532nm；
5. Error oflaser output energy: ≤±20%;
6. Spot size:2-10mm continuously adjustable, error less than ±20%;
7. aiming beam wavelength: 635 nm; output power Pc shall be 0.1mW≤Pc≤5mW;
8. The distance betweenspot center and aiming beam center ≤0.5mm;
|Laser type:||High energy Q-Switched Nd:YAG|
|Certification:||CFDA, TGA, TUV medical CE|
|Pulse width:||8ns; 8ns|
|Pulse energy:||1000mJ; 500mJ|
|Pulse energy Range:||100~1000mJ; 100~500mJ|
|Spot size:||2~10mm adjustable|